Quasar K uses the same state-of-the-art CNC lathes and multi-aspheric technology as Quasar and applies it to the keratoconic cornea. A truly junctionless, continuous curve is produced in one smooth cut of the computer-controlled lathe, from the lens centre right to its edge.
Conventional spherical or aspheric lens designs are totally unsuitable for keratoconic eyes. They have inadequate edge-lift to properly align with the abnormal cornea without causing mid-peripheral compression. Existing specific designs for keratoconic patients use either spherical multi-curve technology or high eccentricity conic sections. Both of these have disadvantages.
Spherical lenses with large differences in peripheral curve radii are able to cope with the rapid flattening of the keratoconic cornea from the abnormal centre to the normal periphery, but have very pronounced junctions, which can be a source of discomfort and mechanical insult. Aspheric lenses with rapidly flattening elliptical surfaces have smoother back surfaces, but give rise to induced off-axis astigmatism and poorer visual acuity.
Quasar K uses the same advanced CNC lathes and multi-aspheric technology as Quasar but applies them to the keratoconic cornea. The high-precision, computer-controlled lathe achieves a truly junctionless, continuous curve from the lens centre to its edge in one single, smooth cut. This surface exploits the clinically validated approach of using a smaller B.O.Z.D. with steeper B.O.Z.R.s and higher axial edge clearances for steeper B.O.Z.R.s.
The intention is that the fitting pattern should be similar over the vast majority of different cases presenting in the consulting room, from early keratoconus to well-advanced disease. Whilst the surface is truly aspheric itís polynomial nature is such that it enables you to combine the comfort of a conventional elliptical lens with the visual advantages of a spherical lens without the attendant disadvantage described above.
Computer modelling and consistent results have validated this approach and demonstrated its superiority in terms of the visual result and the improvement of patient comfort.
The optimally fitting lens in keratoconus is only loosely linked to the keratometry readings. Therefore, careful assessment of lens fit with fluorescein is crucial to success. However, an appropriate starting point in early keratoconus will be made by choosing a B.O.Z.R. close to the flattest keratometry reading. As the disease progresses, keratometry readings become increasingly meaningless, but where readings are attainable, a starting point will be the average of the flattest and steepest reading.
It must be emphasised that there is not a substitute for careful lens observation and several trial lenses will probably be required until the practitioner is entirely happy with the fitting.
The table below is for guidance only:
|Typical K Reading mm
|6.80 – 7.80
|= Flattest K
|6.00 – 6.80
|5.00 – 6.00
Desirable Fitting Patterns
Always assess the fitting with the lens centred on the cornea.
Always allow sufficient time for the lens to equilibrate on the eye. Instil fluorescein and ensure that the patient blinks sufficiently for it to penetrate under the central portion of the lens. A well-fitted Quasar K should show 2-3mm of light apical touch. This should be complemented by mid-peripheral touch, as shown in Fig 1. In-between the two areas of touch there will be an annulus of fluorescein, which should be free of bubbles. If no central touch is seen, as seen in Fig 2, then the B.O.Z.R. should be flattened until the correct appearance is seen. Conversely, if there is excessive central touch, as shown in Fig 3, then the B.O.Z.R. should be steepened until only a light touch is visualised.
Figure 1 – Mid-peripheral Touch
Figure 2 – No Central Touch
Figure 3 – Excessive Central Touch
If any part of the lens edge shows
Centration is particularly difficult to control in keratoconus. About 75% of cones are displaced down with regard to the pupil centre and therefore a well-fitted lens will usually centre slightly downwards. A high-riding lens is often too flat or has too great an edge clearance. To improve the optical result with a low-riding lens, it is best to increase the diameter if in other respects the fitting is optimal.
The trial lenses are increasingly negatively powered as the B.O.Z.R. increases. This is to help reduce the magnitude of the over-refraction in more advanced cases. There will, in many cases, be a small residual cylinder, but wherever possible try and optimise the vision using spherical lenses only. Ensure the over-refraction takes place using a trial lens of the same B.O.Z.R. as the final lens as small changes to the B.O.Z.R. can dramatically affect the visual acuity and over-refraction. Ensure that the initial lacrimation has subsided. Obviously, the vertex distance should be taken
Where there is a marked residual cylinder (>0.75) try to dispense
- Junctionless back surface
- Improved centration
- Improved comfort
- Comprehensive trial sets
- Unparalleled technical support
- Choice of lens materials
|Trial Set Radii
|5.70 to 8.00 in 0.10mm steps
|-9.50 to – 3.00D, radius and overall diameter dependent
|Total Axial Edge Lift
|0.421mm – 0.196mm, radius and overall diameter dependent
|8.80 to 9.60mm radius dependent
Meet every challenge with the Quasar range of RGP lenses. Range includes: Single Vision, Toric and Multifocal.
Contact Lens Wear Without Compromise
Using breakthrough manufacturing technology, Tangible Hydra-PEG bonds a 40nm hydrophilic coating to the surface of specialist contact lenses, significantly increasing wettability and lubricity.
The result? Happy patients who can enjoy a life of comfortable contact lens wear free from compromise.
Tangible Hydra-PEG is available with all No7 GP lenses. Just ask our Customer Care team to add Tangible Hydra-PEG when you place your next order.
Perfect for all GP patients, Tangible Hydra-PEG offers:
Increased wearing times
Improved ocular health
Superior surface properties
Improved wearing experience
“Brilliant! I’ve gone from using eye drops 5 times a day to 0...”
Keratoconus RGP wearer | 50+ years
Research shows that 70% of patients prefer a Tangible Hydra-PEG coated lens to their habitual, uncoated lens.
End of day comfort
67% of patients said they prefer a lens coated with Tangible Hydra-PEG than uncoated.
Dry eye comfort
When asked about comfort, dry eye patients rated lenses coated with Tangible Hydra-PEG ‘8 out of 10’, compared to ‘3 out of 10’ for uncoated.
70% prefer a Tangible Hydra-PEG coated lens
No7 Recommends - Ote Fine
All-in-one comfort solution for RGP contact lenses
Manufactured using the highest grade ingredients and exclusive to independent opticians, Oté Fine is the complete solution for all your RGP patients. Oté Fine is perfect for your EyeDream Ortho-k wearers too.
We have a dedicated technical support team, headed by Kevin, with decades of experience ready to help and advise you on any fitting issues. If you need support, please call:
Frequently Asked Questions
We’re here to help. If you can’t find the answer to your question, why not give us a call. Our offices are open Mon-Fri 9:00 – 17:30
Can I order these lenses empirically
How long will it take to manufacture lenses
What is the returns policy?
We are here to help
Drop us a line anytime, and one of our customer service team will respond to you as soon as possible.