KeraKone is a rigid contact lens for keratoconus that takes advantage of a well-established approach to keratoconic lens fitting through adaptive design and optic zone control.
KeraKone is a multi-spherical back surface Keratoconus lens and takes advantage of a well-established approach to keratoconic lens fitting through adaptive design and optic zone control.
KeraKone has three standard peripheral systems – STANDARD, INCREASED and DECREASED. These three options should fit the majority of keratoconic patients.
Aims and Ideal Fit
- The aim of fitting KeraKone is to fit a lens which provides good acuity whilst being comfortable and maintaining corneal integrity.
- The ideal fit (Fig. 1) will have:
- Light feather touch at the apex of the cone
- A small amount of apical clearance is acceptable
- As close to alignment as possible over the cornea
- Edge clearance producing a fluorescein band of 0.5 to 0.8mm.
- Good centration
Figure 1 – Ideal Fit
Trial Lens Selection
- Topography fitting:
- Start with a trial lens which has a base curve close to the measurement at the temporal 4mm position
- Keratometer fitting:
- Start with a trial lens 0.20mm steeper than mean K
Fitting and Troubleshooting
- We recommend the use of a short-acting topical anaesthetic to reduce tearing and enable accurate fluorescein assessment.
- After instilling fluorescein, wait at least 2 minutes for the excess fluorescein to dissipate before assessing the fit, or a false fitting pattern may be observed.
- Central fit assessment:
- Aim for light feather touch at the apex.
- Excessive apical clearance (Fig. 2) may result in reduced visual acuity; flatten the base curve until very light apical touch can be discerned.
- Excessive apical bearing (Fig. 3) increases the risk of corneal abrasion and scarring; steepen the base curve until light feather touch is seen.
Figure 2 – Excessive Apical Clearance
Figure 2 – Excessive Apical Bearing
Peripheral fit assessment:
- There will usually be some
toricityto the edge clearance so it should be assessed along both the flat and steep meridians.
- Excessive edge lift (Fig. 4) will result in a lens which moves too freely.
- Inadequate edge clearance (Fig. 5) will be seen as a fluorescein band of less than 0.5mm and a dark ring of touch near the edge of the lens. The lens may be uncomfortable and may not move freely.
Edge profile adjustments:
- Fitting sets are supplied with standard edge lift
- 2 steps of increased and 2 steps of decreased edge lifts are available
- Each increment has been calculated to produce a clear and noticeable change in the fit
- The standard diameter is 8.7
- Lens diameter should only need to be altered in advanced cones or very toric corneas
- If the lens is riding low, increase the total diameter
- If the lens is riding high, decrease the total diameter
- Over-refract initially using 1D steps and then refine using 0.50D and 0.25D steps
Figure 3 – Excessive Edge Lift
Figure 4 – Inadequate Edge Clearance
Fig. 4 and Fig. 5 courtesy of Keith Tempany FBDO CL FBCLA.
|Design||Number of Lenses||BCOR||Diameter||Power|
|KeraKone||24||5.10 to 7.40mm (0.10mm steps)||8.70mm||-23.00 to -2.00D (in 1.00D steps)|
Why KeraKone 2?
- Suitable for all keratoconic profiles
- Aberration control optics providing excellent visual acuity
- Reduced flare and glare
- Extensive base curve and diameter range
- Familiar fitting strategy
Firstfit for irregular corneas such as KC and Post Lasik Ectasia
- Refits from other leading designs
- When soft KC lenses are no longer adequate
Optimum and Boston
|Dk||Dependent on material|
|Tint||Dependent on material|
|Base Curve||4.50 to 7.50mm in 0.05mm steps|
|Diameter||8.70mm (standard), 8.10mm, 8.40mm, 9.00mm, 9.30mm|
|BVP||+25.00D to -25.00D|
|Edge Profile||Standard, 2 steps available flatter and steeper|
Contact Lens Wear Without Compromise
Using breakthrough manufacturing technology, Tangible Hydra-PEG bonds a 40nm hydrophilic coating to the surface of specialist contact lenses, significantly increasing wettability and lubricity.
The result? Happy patients who can enjoy a life of comfortable contact lens wear free from compromise.
Tangible Hydra-PEG is available with all No7 GP lenses. Just ask our Customer Care team to add Tangible Hydra-PEG when you place your next order.
Perfect for all GP patients, Tangible Hydra-PEG offers:
Increased wearing times
Improved ocular health
Superior surface properties
Improved wearing experience
“Brilliant! I’ve gone from using eye drops 5 times a day to 0...”
Keratoconus RGP wearer | 50+ years
Research shows that 70% of patients prefer a Tangible Hydra-PEG coated lens to their habitual, uncoated lens.
End of day comfort
67% of patients said they prefer a lens coated with Tangible Hydra-PEG than uncoated.
Dry eye comfort
When asked about comfort, dry eye patients rated lenses coated with Tangible Hydra-PEG ‘8 out of 10’, compared to ‘3 out of 10’ for uncoated.
70% prefer a Tangible Hydra-PEG coated lens
Tangible Hydra-PEG Downloads
No7 Recommends - Ote Fine
All-in-one comfort solution for RGP contact lenses
Manufactured using the highest grade ingredients and exclusive to independent opticians, Oté Fine is the complete solution for all your RGP patients. Oté Fine is perfect for your EyeDream Ortho-k wearers too.
We have a dedicated technical support team, headed by Kevin, with decades of experience ready to help and advise you on any fitting issues. If you need support, please call:
Frequently Asked Questions
We’re here to help. If you can’t find the answer to your question, why not give us a call. Our offices are open Mon-Fri 9:00 – 17:30
Can I order these lenses empirically
How long will it take to manufacture lenses
What is the returns policy?
We are here to help
Drop us a line anytime, and one of our customer service team will respond to you as soon as possible.